Application fields:Composition and inspection of ordinary abrasive particle size and production and grading of super hard materials.
Applicable materials:Geological, metallurgical, powder, chemical, construction, cement, medicine, national defense and other departments of scientific research and production, as well as the laboratory for screening and testing of materials.
Slap type laboratory vibrating screen is one of commonly used equipment in the laboratory,it can be pided into belt and gear slap type laboratory vibrating screen. Its structure is mainly composed of motor base,sieve set and transmission mechanism,which be equipped with special fixture,not only can assemble Φ200 standard test sieves,but can assemble sieve set of Φ 75,Φ100 and Φ300.It can be handling with convenient and flexible,and can automatically shut down.
1. Simple structure, long service life and low maintenance cost.
2. The transmission is accurate, the consistency is good, The gear drives the synchronous belt in a way that the speed ratio is precise,and no speed drop.
3.High screening efficiency and unique design makes slap screen at work and also have two forms of movement : reciprocating and clapping vibrations,
so there will be no blockage of material and the problems and fast cutting, screening thoroughly, high efficiency.
4.The timing function ensures that the inspection results are available to the screen power controller for the time of reservation, and the automatic shutdown inspection and production are consistent.
Slap type laboratory vibrating screen (belt slap type laboratory vibrating screen./gear slap type laboratory vibrating screen)Driven by a motor through a synchronous belt drive gear to the spindle, After the gear is changed, the spindle rotates according to the required speed. The spindle drives the eccentric wheel to make the skateboard reciprocate and drive the slap type screen installed in the skateboard tray to shake left and right. At the same time, the spindle drives the gear rotation of the ejector rod. After changing speed, the ejector gear hits the ejector rod through the cam, and the ejector makes the hammer beat up and down on the upper cover of the test screen, which makes the material in the test screen and the sample sieve reciprocate and vibrate up and down.The material particles smaller than the standard screen mesh are screened into the lower test screen, so that only the same size material is left in each test screen. To achieve the separation of different size and determine the size of this material, finally the material in the test and sample sieving is screened.
|3||200 test sieve||set||1|